This is the project used to identify and measure unknown resistance, capacitance and inductance. When we connect any resistor or capacitor or inductor in the breadboard circuit will find whether it is a resistor or capacitor or inductor and calculate the exact value of the resistance or capacitance or inductance and it is displayed in the 16×2 LCD. This is done by utilizing the ADC feature of the Arduino and the Auto ranging voltage divider concept to measure the resistance with reduced errors and RC charging circuit to develop the Capacitance meter. Now-a-days we use multimeter with range selector switches to find the unknown components which will consume time, so we also introduced Auto ranging feature to reduce the manual ranging functions by implementing logics in coding and hardware which is discussed below.
The entire project can be divided into three basic blocks;
Auto ranging Resistance Sensor Unit
RC Charging circuit
Resonating tank circuit
Fig. 1: Block Diagram of Arduino based RLC Meter
Auto ranging Resistance Sensor Unit:
A basic voltage divider circuit is used as the Resistance Sensing Unit to provide an output, which is the voltage equivalent of the unknown resistor as the input. From this output voltage we can calculate the value of unknown resistance, detect the diode type or detect a continuous path. For more details you can refer here.
A scheme for estimating the value of unknown resistance value roughly and then putting a matching resistor in place of R2 is what we need here and this method is called auto ranging. The circuit given below demonstrates auto ranging. For more details you can refer here.
RC Charging circuit:
The Capacitance sensor in this project is an RC discharging circuit in which the unknown capacitor is discharged through a known resistor. An RC discharging circuit is a circuit in which a capacitor and a resistor are connected in parallel. In an RC discharging circuit the time taken for the capacitor to discharge through the resistor to half the voltage before starts discharging is dependent on the values of resistance and capacitance. For more details refer this link.
Resonating tank circuit:
This project makes use of the resonance frequency of a tank circuit, in which there will be a capacitor and an unknown inductance connected in parallel. The natural resonating frequency of the tank circuit varies with the value of the inductor connected. The frequency is measured and the value of inductance is calculated with the help of simple mathematical equations. For more details refer this link.
Digital pin 9 is used to isolate resistance and capacitance sensing circuit by means of proving base signal to the npn transistor, which is used to isolate 100k resistor while measuring resistance and adding 100k while measuring capacitance. This circuit isolate 100k resistor which is used to create RC circuit from the auto ranging voltage divider circuit to avoid mixing up of these sensing circuits.
The processor unit in this project is the Arduino board and it uses the ADC module to read the output voltage from the Sensor Unit. On the Arduino board, we are using an 8 channel, 10 bit ADC with the reference voltage pin connected to 5 V. The ADC reads the voltage V2 and generates an equivalent value ‘ValueADC‘at the ADC register. For more details refer this link.
The display unit in this project uses a common 16*2 LCD on which the Arduino displays the resistor value and diode type. The LCD has been wired in four bit mode to reduce the number of output pins of the Arduino board to be used. The code running in the Arduino used the library function analogRead() to obtain the ADC value and lcd.print() to display the data on 16*2 LCD.
Fig. 2: Prototype of Arduino based Electronic Component Identifier designed on a breadboard